Digital early pregnancy test monitorsDelivery term:The date of payment from buyers deliver within days
- Total supply:
- Delivery term:
The date of payment from buyers deliver within days
- Validity to:
- Last update:
- Browse the number:
- Kantech International (Jiangxi) Co. Ltd By certification [File Integrity]
698 Jing Dong Avenue, Nanchang, Jiangxi, China (330029)
- Website: http://kantecare.hongxunpipe.com/
One step digital pregnancy test
A multi-center clinical evaluation was conducted comparing the digital pregnancy test and another commercially available urine membrane hCG test. The urine study included 100 specimens and both assays identified 60 negative and 40 positive results. The results demonstrated a 100% overall agreement (for an accuracy of > 99%) of the digital pregnancy test when compared to the other urine membrane hCG test.
Sensitivity and specificity
The digital pregnancy test detects hCG at concentrations of 25mIU/ml or greater. The test has been standardized to the W.H.O. Third International Standard. The addition of LH (500 mIU/ml), FSH (1000 mIU/ml), and TSH (1000 μIU/ml) showed no cross-reactivity to negative (0 mIU/ml hCG) and positive (25 mIU/ml hCG) specimens.
Product specification (for in-vitro diagnostic use)
Digital early pregnancy test is >99% accurate at detecting the pregnancy hormone (hCG). Results take 5minutes.
Digital ovulation test is >99% accurate at detecting the ovulation hormone(LH). Results take 5 minutes.
Decisive Yes/No readout
Easy to use
- FOB Port: Shanghai,Ningbo,Shenzhen
- Lead Time: 15 - 30 days
- Dimensions per Unit: 18 × 5 × 2.5 Centimeters
- Weight per Unit: 45 Grams
- Units per Export Carton: 500
- Export Carton Dimensions L/W/H: 50 × 50 × 50 Centimeters
- Export Carton Weight: 17 Kilograms
- Eastern Europe
- North America
- Mid East/Africa
- Central/South America
- Western Europe